We have developed a highly automated IRAF-based data reduction package, CIRDR, that can be used for both quick look data inspection, and generation of astrometrically calibrated science quality 40964096 images based on a contiguous 44 mosaic.
CIRSI, the Cambridge Infra-Red Survey Instrument, is a new near-infrared camera that was built at the Institute of Astronomy. It consists of four 10241024 pixel Rockwell HAWAII HgCdTe (Mercury-Cadmium-Telluride) detectors, arranged in a sparse-filled mosaic--making it the largest infrared camera in the world to date. CIRSI's unprecedented wide field of view means it is ideally suited for survey work. The previous generation of infrared detectors (typically 256256 pixels) tended to have too limited a field of view for panoramic surveys. This was unfortunate, as the near-infrared is very desirable for such survey work. For example, when searching for high-redshift clusters of galaxies in optical wave-bands, one finds that beyond about , the contrast against the very high field galaxy counts makes detections less and less reliable. Additionally, the k-correction tends to dim the light of distant early-type galaxies, which kills the cluster contrast at optical wavelengths.
Thus we are able to combine the advantages of searching in the near-infrared wave-bands and CIRSI's ability to cover a large solid angle, an essential feature for finding rare objects like rich clusters of galaxies.
Most of our observations have been carried out on the 2.5m Isaac Newton Telescope on the Canary Islands. So far this has consisted of four observing runs, totaling 35 nights.
Table 1 lists our most important cluster observations.
At a scale of 0.45''/pixel on the INT, an image from a single chip has a 7.7' field of view. Hence a filled 44 mosaic (see Figure 1) is just over 30'30' in size.
The camera is typically read out in non-destructive read mode (NDR)--see Figure 2 for an illustration.
The controller is a standard LSR-AstroCam 4100 CCD controller. Each quadrant on a chip is read out individually, giving images for each read. Hence, for a typical observation, which would be a 40-second exposure consisting of 3 reads, the data rate would be
|0.5MB||[size of a 512512 16-bit quadrant image]|
|[number of quadrants]|
|[number of reads]|
giving 24MB in 40 seconds, i.e. 36MB a minute which comes to over 2GB an hour at a sustained rate.
Each observation pointing is then repeated, offset by a small amount (typically 15''). Usually 9 or so of these dithers then make up a pointing.
In order to deal with such large volumes of data, we employed an observing system consisting of several computers and running a variety of operating systems, linked by fast 100Mb Ethernet. Storage was provided by several 16GB hot-swappable hard drives, in addition to multiple SCSI disks on the individual machines. Data would be written to DDS-3 tapes during the daytime.
The user interface is provided by a GUI, a program called PixCel, running on a Microsoft Windows 95 PC. The raw data is copied across automatically to Linux machines for reduction.
We have developed CIRDR, a highly automated IRAF-based piece of software, for reducing our data. This was initially intended as a quick look data inspection facility, but has since evolved into a fully-fledged data reduction package. The requirements for such a package are manifold:
For these reasons, and to maintain backward compatibility with previous software, we chose to base our package in IRAF. CIRDR is essentially a collection of CL scripts, with certain (speed-critical) tasks written in SPP.
The multiple tasks required for reducing are as follows:
Even though these steps are all performed by individual sub-packages and tasks (which has the advantage that they are also usable individually, and could be re-used in separate packages), there exists a monolithic blanket task which runs through a large part of them fully automatically after initial input from the user.
Since earlier this year, with the start of CIRSI observations at the 2.5m DuPont Telescope in Chile, things have changed somewhat. Data is now read out in read-reset-read mode (RRR), and the images are written to disk pre-assembled and reset-corrected. Sky subtraction (and/or flat-fielding) is performed, after which the objects are located and subsequently matched with the on-line APM catalogue. This gives the exact offsets for co-addition, and for mosaicing if it is required. This new version of CIRDR is now written in a hybrid of CL and C.